Looking for a high-quality, clinical-lab grade device to perform reliable, real-time quantitative PCR tests? Trying to set up a small PCR test lab but being constrained by the high cost of a high-quality PCR testing device? Look no further! Aurora is glad to present you with its compact and budget-friendly QPCR device – “UltraFast (FQ-8A) ”. This machine is ideal for low-throughput real-time QPCR testing with 8 sample slots per run. Each run takes around 30 minutes and yields a high standard result, no matter if it is for research use or clinical diagnosis.
- Single machine operation, easy to operate
- Light and compact, easy to carry
- Efficient temperature control system, the maximum temperature change rate can reach 8 ℃ per second
- Compatible with the Coronavirus Nucleic Acid Extraction kits of Biogerm, Sansure, Shanghai ZJ Bio-Tech , maccura, Aurora, etc.
Run the program
Choose your test
Run time: ~30 mins
|Run time for 40-cycles PCR reaction
|Average heating-up rate
|50℃ – 90℃，≥3.0℃/s
|Maximum heating-up rate
|50℃ – 90℃，≥6.0℃/s
|Average cooling rate
|90℃ – 50 ℃，≥3.5℃/s
|Maximum cooling rate
|90℃ – 50℃，≥6.5℃/s
|Module control accuracy
|Absolute value of difference between measured value and set temperature ≤0.5℃
|Temperature Accuracy during the run
|The CV of the Ct value shall not be greater than 3%
|Linear regression coefficient r≥0.990
|Protection level of equipment shell
|Dimensions (in mm)
|50L × 190W × 120H
|3.2kg (With power adapter 3.8kg)
|3.2kg (Including power adapter 3.8kg)
|25cm x 19cm x 12cm
|Automatic PCR amplification and result output
|4 monochrome LED
|Max Temperature rate
|Detectable more than 200copies/ml
What is the compatible sample reaction tube/plate to the UltraFast QPCR (FQ-8A)?
The device only run 8 samples in capped striped/non-striped PCR 0.1 ml transparent thin-walled tubes.
How do UltraFast QPCR (FQ-8A) machines compare to a regular thermal cycler?
UltraFast QPCR is a compact (small), budget-friendly device with ease of operation for real-time PCR (quantitative and qualitative results up to 8 samples.
Does the compact UltraFast QPCR (FQ-8A) instrument use standard PCR consumables and reagents?
UltraFast QPCR (FQ-8A) reagents are similar. But the consumables are only striped/non-striped PCR 0.1 ml transparent thin-walled tubes and 57 x 30 mm receipt paper.
Does the UltraFast QPCR (FQ-8A) need to be connected to a computer or laptop to run?
No. The compact UltraFast QPCR (FQ-8A) does not need any external devices and has its ow7-inch touch screen. Users can program and operate the device using the built-in software.
How can we extract the result from the UltraFast QPCR (FQ-8A)?
The results as graphs and values can be seen on the screen and could also be collected using a USB flash drive and also the raw values can be printed right away on a receipt paper from the device.
What are the examples of application sectors for UltraFast QPCR?
Rapid detection of virus nucleic acid, small size, lightweight, can be applied in the following scenarios: Research Labs, Hospital POCT, CDC, Customs, Airport Quarantine, Primary Medical Institutions, field emergency detection, animal quarantine, food safety, etc.